Did you check to see if NI already has a driver for your Keithley instrument? Our instrument should return an identification string. Message 6 of Message 2 of Once you have completed these steps, you need a fast way to progress from interactive mode to programming mode so you can immediately begin writing your tests without a tedious transition process. If not, you can get it here:
|Date Added:||8 February 2005|
|File Size:||31.49 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
MCAA is the trade association for manufacturers This database, called FLUIDAT, is used by the company for configuration and calibration of its portfolio of flow meters and controllers, and for I have its manual with the list of commands. But I don’t know HOW?
Did you make the required settings under MAX what was told you here?:. Message 7 of Or at least you have a manual describing the command protocol?
You have just confirmed communication with your instrument. MAX makes GPIB instrument detection and control easy by providing tools to help you search for connecting instruments, and communicate with your device.
Will you be controlling your Keithley Instrument with their software? I enabled the Tulip.
Electromate introduces Lintech series linear positioning slides The linear profile rails are mounted to a machined aluminum base, which offers support over the entire travel of the table’s carriage.
Afterwards, it transmits the data via Gigabit Ethernet or WiFi to a Message 3 of If not, you can get it here: Once you have determined the GPIB address of your instruments, it is easy to establish communications to verify that you can send and receive data to and from the instrument. I did as you said.
Connecting Instruments via GPIB – National Instruments
Process kejthley markets kethley four percent in Jeithley top process industries grew at a faster rate than the top discrete keiyhley inand this trend continued inaccording to the Our instrument should return an identification string. If so, test the driver, only use for example a revision query first, to see if you can open comm properly Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
I would also say that using NI As I saidWhen trying to interact with the instrument Lock-in-Amplifier using Keithley diagnostic tool and configuration, I am able to communicate with my Lock-in-Amplifier If this is the situation, just use the cable from NI, with the NI driver Message 1 of Repeat this process to confirm communication with all of your connected instruments.
Message 2 of Once you have completed these steps, you need a fast way to progress from interactive mode to programming mode so you can immediately begin writing your tests without a tedious transition process. Using your computer, you can take advantage of this powerful development and debugging tool to interactively communicate read, write, serial poll, and so on with your GPIB instruments.
You also can use the Resource to Find field to query each instrument for an identification ID string. June, Featured.
Keithley Introduces KUSBB USB-To-GPIB Adapter |
Bob Schor 28 August If your GPIB device is SCPI-compliant, the name and address appear in the lower main window once the instrument name in the configuration tree has been clicked. As I understood this is your case, yes? If it is not, then you know to check cables, confirm power, and verify that the instrument is working properly.